Documents Input modules You can connect up to four 4 to 20 mA transmitters with 2 or 3-core cables to an input module. Up to a total of four input modules per controller can be installed. Each module is populated with three relays. In addition, the two alarm levels are fully configurable and can be set as rising or falling and latching or non-latching. A third alarm is possible by using the relay module. Each module is populated with eight relays.
|Published (Last):||18 May 2016|
|PDF File Size:||2.80 Mb|
|ePub File Size:||6.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Complex alarms can be fitted to suit your individual requirements, and switching delays can be configured to optimise your processes. The modular structure of the REGARD makes it possible to adapt the design of the system to suit your exact needs. The system can be altered or extended with ease. The advantages of the REGARD in terms of overview and documentation can therefore be translated to the entire system. This prevents the entire system from failing if one component fails single point of failure.
It also makes it easier to add on independent subsystems. The use of optimised software filters in signal preparation, the option of suppressing the alarm and the comparison of analogue and digital transmitted measurement values mean that false alarms are prevented more effectively than ever before.
Special signals errors, warnings, etc. The logical restrictions on the possible manual settings effectively prevent incorrect configurations. A simple menu structure and user-friendly displays on the dashboard, as well as the easy-to-learn operation and the associated symbols, ensure that operation is as safe and secure as possible. You can prepare the configuration via PC software offline and upload it to the system later.
The wiring can be clearly structured and tidied away before the main components are installed. This ensures better preparation of maintenance work with respect to the provision of materials and tools.
It also makes it easier to guide service personnel with greater accuracy. The documentation frequently required by monitoring institutions, for example, can also be generated by the REGARD This documentation is made available without the need for additional tools.
Images and Videos
You can configure up to 16 measuring channels. In addition, the modular setup enables you to customise the control units to the demands of your plant. You can also embed further features to existing alarms. Compare Input modules You can connect up to four 4 to 20 mA transmitters with 2 or 3-core cables to an input module.
Dräger Regard 3900
360 Product View